Types of Implant Surgical Guides in Dentistry: A Review
Various techniques have been proposed for the fabrication of surgical guide templates in implant dentistry. The objective of this paper is to review the associated literature and recent advancements in this field, based on design concept. An electronic and hand search of the literature revealed 3 categories, namely, nonlimiting, partially limiting, and completely limiting design. Most clinicians still adopt the partially limiting design due to its cost-effectiveness and credibility. Moreover, clinicians use cross-sectional imaging during the preimplant assessment of surgical sites.
Key Words: implant guidance, implant placement, surgical guide, surgical template, implant dentistry.
The studies recently on the clinical success of dental implants have indicated a high implant survival rate.
1. Nevertheless, the inadvertent association of most surgical and prosthetic complications with improper diagnosis and implant placement has also been documented.
2. These factors play a crucial role in the long-term predictability and success of implant prosthetics. Surgical guide templates not only assist in diagnosis and treatment planning but also facilitate proper positioning and angulation of the implants in the bone.
3 Moreover, restoration driven implant placement accomplished with a surgical guide template can decrease clinical and laboratory complications.
4. Hence, increasing demand for dental implants has resulted in the development of newer and advanced techniques for the fabrication of these templates.
PARTIALLY LIMITING DESIGN
In such designs, the first drill used for the osteotomy is directed using the
surgical guide, and the remainder of the osteotomy and implant placement is then
finished freehand by the surgeon.6 Techniques based on this design concept
involve fabrication of a radiographic template, which is then converted into a
surgical guide template following radiographic evaluation. Various authors have
proposed different techniques involving modifications in the following stages of
fabrication, namely, material used for the fabrication of the surgical template,
radiographic marker used, type of imaging system used, and the conversion
process involved in converting the radiographic template into a surgical
template. These various techniques are discussed in the Table. Nonetheless, all
of the aforementioned techniques failed to completely restrict the angulation of
the surgical drills.
COMPLETELY LIMITING DESIGN
Completely limiting design restricts all of the instruments used for the osteotomy in a buccolingual and mesiodistal plane. Moreover, the addition of drill stops limits the depth of the preparation, and thus, the positioning of the prosthetic table of the implant. As the surgical guides become more restrictive, less of the decision-making and subsequent surgical execution is done intraoperatively. This includes 2 popular designs: cast-based guided surgical guide and computer-assisted design and manufacturing (CAD/CAM) based surgical guide.
Cast-based Guided Surgical Guide
The surgical guide is a combination of an analog technique done along with bone sounding and the use of periapical radiographs in a conventional flapless guided implant surgery.6 The periapical radiograph is modified using digital software to help in transposition of root structure onto the cast. The cast is then sectioned at the proposed implant site, and bone-sounding measurements are transferred to help in orientation of the drill bit to perform a cast osteotomy. A laboratory analog is placed in the site, and a guide sleeve consistent with the implant width is modified using wires that are used to create a framework around the teeth. Vinyl polysiloxane occlusal registration material is used to form the superstructure.
CAD/CAM-BASED SURGICAL GUIDE
CAD/CAM technology uses data from computerized tomography scan (CT)33 to plan
implant rehabilitation. The CT images are converted into data that are
recognized by a CT imaging and planning software. This software then transfers
this presurgical plan to the surgery site using stereolithographic drill
guides.34 CAD/CAM-based surgical guides offer many advantages. For example, the
virtual 3-dimensional (3D) views of the bony morphology allow the surgeon to
visualize the surgical bone site
prior to implant placement; risks such as inadequate osseous support or compromise of important anatomic structures are avoided; incorporation of prosthetic planning using a scanographic template allows the treatment to be optimized from a
prosthodontics and biomechanical point of view35; and the technique promotes flapless surgeries, allows presurgical construction of the master cast and provisional restorations, and facilitates immediate loading.36 Accuracy of CAD/CAM technology in dental implant planning and predictable transfer of the presurgical plan to the surgical site has been documented.37–46 However, the effectiveness has not yet become an established fact and still needs ongoing research. This technique has certain drawbacks. Special training for familiarity with the entire system and special equipment is necessary.
Also, a considerable number of technique-related complications were observed. The various complications47 recorded were related to inaccurate planning, radiographic stent error, intrinsic errors during scanning, software planning, the rapid prototyping of the guide stent, and the transfer of information for the prosthetics. However, if the clinician recognizes these sources of inaccuracy, efforts can be made to minimize the error and optimize patient treatment. The procedure for fabrication of CAD/CAM based surgical guides can be divided into the following steps:
1. Fabrication of the radiographic template,
2. The computerized tomography scan,
3. Implant planning using interactive implant surgical planning software &
4. Fabrication of the stereolithographic drill guide.
The radiographic template must be an exact replica of the desired prosthetic end
result, as it allows the clinician to visualize the location of planned implants
from an esthetic and biomechanical standpoint.48 This is followed by fabrication
of an interocclusal index, to allow reproducible placement of the scan template
intraorally.49 A double scanning procedure is then followed.49 The patient is
scanned wearing the radiographic scan template and radiographic index (interocclusal
index) during the first scan, whereas the second scan is performed without the
index. The first scan is used to visualize the bony architecture and anatomy of
the site of interest, and a second scan is performed to visualize the
guide. The 2 resulting sets of 2D CT data (Digital Imaging and Communication in Medicine [DICOM files]) are then superimposed over each other according to the radiographic markers and are further converted into a file format compatible with the 3D planning program.49 Resulting from this fusion is an exact representation of the patient’s bone structure and scanning denture in 3D space. At this point, the virtual surgical procedure can be performed.49,50 A 3D implant planning software allows for simultaneous observation of both the arches and the radiographic scan template in 3 spatial planes and helps to virtually plan the location, angle, depth, and diameter of the virtual implants. It produces an axial image, a panoramic image, and a series of cross-sectional images on the screen at the same time. Various implant planning software products are available commercially,
namely, SimPlant, SurgiCase (Materialise Dental Inc, Leuven, Belgium), Procera (Nobel Biocare, Go¨teborg, Sweden), ImplantMaster (I-Dent Imaging Ltd, Hod Hasharon, Israel), coDiagnostiX (IVS Solutions AG, Chemnitz, Germany), and Easy Guide (Keystone Dental, Burlington, MA). Once the computer planning is accomplished, this plan is saved as a ‘‘.sim’’ file and sent to the processing center for fabrication of the surgical guide, using stereolithography. Stereolithography34 is a computer-guided, laser-dependent, rapid prototyping polymerization process that can duplicate the exact shape of the patient’s skeletal anatomic landmarks in a sequential layer of a special polymer to produce a special 3D transparent resin model, which fits intimately with the hard and/or soft tissue surface. Once hardened, the polymeric prototype contains spaces for stainless steel or titanium drill-guiding tubes. These tubes precisely guide the osteotomy drills, precluding the need for the pilot drills.
Although the completely limiting design is considered a far superior design
concept, most clinicians still adopt the partially limiting design due to its
cost-effectiveness and credibility in the field. In addition, it has been
observed that most clinicians
use surgical guide templates that are based on cross-sectional imaging to facilitate accurate planning and guidance during the surgical phase. Evidence-based research still needs to be conducted to evaluate the applications of the completely limiting design and its effect on the treatment outcome in oral implant logy.
Data (3D Implant Surgical Guide) - access through http://www.joionline.org
Data images downloaded - http://www.google.com
Safety Regulations | Notices | Wieland Post Arts | Zenostar | Digital Impression
Digital Dentistry | CNC Milling Centre | 3D Printing Model | 3D Implant Surgical Template
Zenotec Milling | Videos | Materials | Labsheet - Work Form
This site was last updated Friday, 10. November 2017
Working Hours: Monday to Friday 0830hrs - 1730hrs
Saturday to 0830hrs - 1230hrs
Sunday & Public Holiday - Closed
ORIENT DENTAL LABORATORIES & SUPPLIERS PTE LTD. (S)INGAPORE
NO. 33 UBI AVENUE 3 #05-51 VERTEX TOWER BUILDING A
TEL: +65 67483113 | +65 67483685 | FAX: +65 67481597
email@example.com | firstname.lastname@example.org
Page best viewed with